AffiliationNorth Western Medical Physics Department, Christie Hospital NHS Trust, Withington, Manchester, UK.
MetadataShow full item record
AbstractA prototype water filled ultrasound string test object is presented for performance checks on Doppler ultrasound systems. The string motion produces realistic and reproducible Doppler wave forms on the system by means of a programmable stepper motor. Wave forms can be generated using a computer, and programmed into the device. The test object can be used to check Doppler calculation parameters, such as pulsatility index and resistance index, as well as range gate registration and wall filter function. The test object is water filled, and the lack of attenuation did prove to be a problem on some systems. The object was tested using five different duplex and colour scanners, and found to be sufficiently reproducible for use as a field test object for Doppler ultrasound machines. The device is compact and portable, making it suitable for checks in the field for both longterm monitoring of performance and inter-machine comparisons.
CitationA programmable Doppler string test object. 1993, 38 (11):1623-30 Phys Med Biol
JournalPhysics in Medicine and Biology
- Testing ultrasonic pulsed Doppler instruments with a physiologic string phantom.
- Authors: Phillips DJ, Hossack J, Beach KW, Strandness DE Jr
- Issue date: 1990 Aug
- Development of an example flow test object and comparison of five of these test objects, constructed in various laboratories.
- Authors: Teirlinck CJ, Bezemer RA, Kollmann C, Lubbers J, Hoskins PR, Ramnarine KV, Fish P, Fredeldt KE, Schaarschmidt UG
- Issue date: 1998 Feb
- Validation of a sensitivity performance index test protocol and evaluation of colour Doppler sensitivity for a range of ultrasound scanners.
- Authors: Browne JE, Watson AJ, Hoskins PR, Elliott AT
- Issue date: 2004 Nov
- A review of Doppler ultrasound quality assurance protocols and test devices.
- Authors: Browne JE
- Issue date: 2014 Nov
- Performance tests of Doppler ultrasound equipment with a string phantom.
- Authors: Goldstein A
- Issue date: 1991 Mar