Design and implementation of a head-and-neck phantom for system audit and verification of intensity-modulated radiation therapy.
AffiliationNorth Western Medical Physics, Christie Hospital NHS Foundation Trust, Manchester, U.K. Gareth.Webster@physics.cr.man.ac.uk
MetadataShow full item record
AbstractThe head and neck is a challenging anatomic site for intensity-modulated radiation therapy (IMRT), requiring thorough testing of planning and treatment delivery systems. Ideally, the phantoms used should be anatomically realistic, have radiologic properties identical to those of the tissues concerned, and allow for the use of a variety of devices to verify dose and dose distribution in any target or normaltissue structure. A phantom that approaches the foregoing characteristics has been designed and built; its specific purpose is verification for IMRT treatments in the head-andneck region. This semi-anatomic phantom, HANK, is constructed of Perspex (Imperial Chemical Industries, London, U.K.) and provides for the insertion of heterogeneities simulating air cavities in a range of fixed positions. Chamber inserts are manufactured to incorporate either a standard thimble ionization chamber (0.125 cm3: PTW, Freiburg, Germany) or a smaller PinPoint chamber (0.015 cm3: PTW), and measurements can be made with either chamber in a range of positions throughout the phantom. Coronal films can also be acquired within the phantom, and additional solid blocks of Perspex allow for transverse films to be acquired within the head region. Initial studies using simple conventional head-and-neck plans established the reproducibility of the phantom and the measurement devices to within the setup uncertainty of +/- 0.5 mm. Subsequent verification of 9 clinical head-and-neck IMRT plans demonstrated the efficacy of the phantom in making a range of patient-specific dose measurements in regions of dosimetric and clinical interest. Agreement between measured values and those predicted by the Pinnacle3 treatment planning system (Philips Medical Systems, Andover, MA) was found to be generally good, with a mean error on the calculated dose to each point of +0.2% (range: -4.3% to +2.2%; n = 9) for the primary planning target volume (PTV), -0.1% (range: -1.5% to +2.0%; n = 8) for the nodal PTV, and +0.0% (range: -1.8% to +4.3%, n = 9) for the spinal cord. The suitability of the phantom for measuring combined dose distributions using radiographic film was also evaluated.The phantom has proved to be a valuable tool in the development and implementation of clinical head-and-neck IMRT, allowing for accurate verification of absolute dose and dose distributions in regions of clinical and dosimetric interest.
CitationDesign and implementation of a head-and-neck phantom for system audit and verification of intensity-modulated radiation therapy. 2008, 9 (2):2740 J Appl Clin Med Phys
JournalJournal of Applied Clinical Medical Physics
- Development and evaluation of an end-to-end test for head and neck IMRT with a novel multiple-dosimetric modality phantom.
- Authors: Zakjevskii VV, Knill CS, Rakowski JT, Snyder MG
- Issue date: 2016 Mar 8
- Dosimetric verification of surface and superficial doses for head and neck IMRT with different PTV shrinkage margins.
- Authors: Shiau AC, Lai PL, Liang JA, Shueng PW, Chen WL, Kuan WP
- Issue date: 2011 Mar
- Dosimetric effects of multileaf collimator leaf width on intensity-modulated radiotherapy for head and neck cancer.
- Authors: Hong CS, Ju SG, Kim M, Kim JI, Kim JM, Suh TS, Han Y, Ahn YC, Choi DH, Nam H, Park HC
- Issue date: 2014 Feb
- Phantoms for IMRT dose distribution measurement and treatment verification.
- Authors: Low DA, Gerber RL, Mutic S, Purdy JA
- Issue date: 1998 Mar 15
- Comparison of measured and computed portal dose for IMRT treatment.
- Authors: Cilla S, Viola P, Azario L, Grimaldi L, Craus M, D'Onofrio G, Fidanzio A, Morganti AG, Piermattei A
- Issue date: 2006 Aug 24